The software update 22.214.171.124 and 126.96.36.199 of the Aethalometer AE33 is a CRITICAL software update.
Current software upgrade focuses on Aethalometers AE33 that run previously released SW versions (188.8.131.52 and 184.108.40.206) that were affected due to some software errors.
Please find below a short description of the corrections made in this software update.
The new software version (v 220.127.116.11) is compatible with all production series of Aethalometers with serial number starting S0, S1, … S7 (S0-S7).
The new software version (v 18.104.22.168) is compatible with production series of Aethalometers with serial number starting with S08.
New software version enables up to 4 different angles of measured light scattering data. In previous versions the nephelometer was only possible to connect with the AE33 in case exactly 4 angles were set, while now the user can chose this parameter between 1 and 4. There are some additional minor fixes (like Flow Calibration report) that were added into the ‘reports’ section.
The instrument must be running Firmware version 513 or greater. We recommend updating the firmware to the current FW518, see User’s Group website for details.
If your Aethalometer has a serial number less than 90 or software version less than 1.172, please contact us for further information before updating.
First log in or sign up for a free account at the Magee Scientific website. You can then access information on all our products. Material specific to the AE33 is under the heading Products > AE33 > Software.
Register for a FREE account Magee Scientific User’s group
Follow the news about our work and research. Please meet us at events worldwide and talk to our experts about your topics of interest.